| Abstract|| |
Purpose: To compare whole-body magnetic resonance imaging (WBMRI) performed at 1.5 and 3 T for technical quality, artifacts, and visibility of selected fixed structures. Patients and Methods: 21 children who had WBMRI at both 1.5 and 3 T scanners within a relatively short interval (3-13 months; average-8.6 months) were included. The images were objectively compared with scores from 4 to 1 for five parameters including severity of artifacts; visibility of liver, vertebral column, and marrow in legs; and overall image quality. Inter-observer agreement was calculated using Kendall's coefficient of Concordance (W) and scores were compared using Signed Rank test. Results: There was substantial inter-observer agreement for all five categories at both field strengths. The difference between averages of mean scores of all five parameters for two field strengths was statistically significant (P < 0.05), indicating less artifact, better fixed structure visibility, and overall image quality at 1.5 T as compared to 3 T. However, scores at 3 T were also rated within a good range (around 3) indicating its feasibility for WBMRI in children. Conclusion: WBMRI at 1.5 T has significantly better image quality, fixed structure visibility, and fewer artifacts, as compared to WBMRI at 3 T in children. This difference is unlikely to significantly affect detection of pathology on 3 T WBMRI as the image quality score at 3 T was also within good range.
Keywords: 1.5 T; 3 T; artifacts; children; image quality; whole-body magnetic resonance imaging
|How to cite this article:|
Mohan S, Moineddin R, Chavhan GB. Pediatric whole-body magnetic resonance imaging: Intra-individual comparison of technical quality, artifacts, and fixed structure visibility at 1.5 and 3 T. Indian J Radiol Imaging 2015;25:353-8
|How to cite this URL:|
Mohan S, Moineddin R, Chavhan GB. Pediatric whole-body magnetic resonance imaging: Intra-individual comparison of technical quality, artifacts, and fixed structure visibility at 1.5 and 3 T. Indian J Radiol Imaging [serial online] 2015 [cited 2020 Jan 29];25:353-8. Available from: http://www.ijri.org/text.asp?2015/25/4/353/169448
| Introduction|| |
Whole-body magnetic resonance imaging (WBMRI) is now recognized as an important tool in both diagnosis and follow-up of various oncologic and non-oncologic conditions in children.,,,,,,, WBMRI can be performed at both 1.5 and 3 T magnetic field strengths. There are a few studies comparing these two field strengths in adults for whole-body MR angiography, oxygen-enhanced MRI of lungs, cardiac and coronary imaging,, and abdominal diffusion-weighted imaging (DWI). Comparison of image quality and artifacts of WBMRI performed at 1.5 and 3 T has also been performed in adults. However, to the best of our knowledge, there has been no study comparing the image quality of WBMRI at these two magnetic strengths in children. It would be ideal to compare studies performed on the same patient analyzing the same pathology simultaneously at these two magnetic field strengths in order to avoid bias secondary to the temporal variation in imaged pathology. However, this would not be feasible, especially in children. We attempted to compare the technical quality at the two magnetic field strengths in the same patients performed at different times but at a time interval.
The aim of our study was to retrospectively perform intra-subject comparison of technical quality, artifacts, and the visibility of selected fixed structures between WBMRI coronal short tau inversion recovery (STIR) sequences performed at 1.5 and 3 T.
| Patients and Methods|| |
Institutional Research Ethics Board approval and waiver for consent were obtained for this study.
A list of children who had undergone MRI studies on both 1.5 and 3 T at our institution, a tertiary pediatric referral center, for various indications was obtained from a database that keeps record of all MRI exams. Children who had WBMRI at least once each on our 1.5 and 3 T Philips (Achieva; Philips Medical system, Best, the Netherlands) MRI scanners from January 2008 to April 2011 were included in this study. Selection of 1.5 T versus 3 T scanner for WBMRI is done clinically based on 3 T compatibility (based on certain devices and prosthesis in the body) and availability of the scanner. Of 22 children who fulfilled these criteria, one child with a gap of 22 months between two WBMRI exams was excluded, as this was deemed to be a long time interval which could potentially cause significant changes in the body physique, physiology, and ability of the child to stay still during the exam. Remaining 21 (6 boys, 15 girls; age between 4.6 and 17 years with average age of 12.03 years at the time of first WBMRI) children who formed the final study group had a maximum gap of 13 months between the WBMRI exams. The minimum interval between WBMRI exams was 3 months, with an average interval of 8.6 months. This interval was thought to be not long enough to change significantly physical characteristics of the body in a child, thus allowing fair comparison of WBMRI technical quality at two field strengths.
The indications in 21 children included 4 cases of Li-Fraumeni syndrome More Details, 5 neuroblastoma, 2 retinoblastoma, 3 rhabdomyosarcoma, and 1 each of acinic cell carcinoma, epithelioid hemangioendothelioma, tuberous sclerosis with epithelioid angiomyolipoma, nasopharyngeal undifferentiated sarcoma, Ewing sarcoma, desmoplastic round cell tumor, and metastatic adenocarcinoma of colon. Nineteen children had WBMRI on both 1.5 and 3 T without sedation or anesthesia; remaining two were under general anesthesia for both 1.5 and 3 T scans.
Magnetic resonance imaging technique
WBMRI was performed using 1.5 and 3 T Philips (Achieva; Philips Medical system) MRI scanners equipped with high-performance gradient systems, sliding table platform, and quadrature body coil integrated in the magnet bore obviating the need to reposition the patient with each station. The exam included single STIR coronal sequence at multiple stations covering the entire body from vertex to toes at multiple stations. The images from multiple stations were retrospectively stitched using software (Mobiview) provided by the vendor to obtain whole-body images. There was some overlap between the stations and stitching was done based on table position. The coronal STIR sequence at 1.5 T was performed with the following parameters: Time to repeat (TR) 3000 ms, echo time (TE) 60 ms, inversion time (TI) 165 ms, field of view (FOV) 400–500 mm, matrix 380 × 290, slice thickness of 5 mm with 1 mm gap, number of signal averages 6, echo train length (ETL) 26, and pixel bandwidth 446 Hz. STIR parameters at 3 T included TR 9126 ms, TE 70 ms, TI 230 ms, FOV 400–500 mm, matrix 460 × 360, slice thickness of 6 mm with 1 mm gap, number of signal averages 2, ETL 27, and pixel bandwidth 256 Hz. Approximate time for each station at 1.5 T was 5 min and at 3 T was 4 min. The total scan time varied from 20 to 35 min depending on the height of the child. The acquisition was free breathing, no sagittal images of the spine were acquired, and no contrast or bowel paralysis was used.
WBMRI images were reviewed by a pediatric radiologist (8 years of experience in reading pediatric body MRI) and a pediatric radiology fellow (A year of experience in reading pediatric body MRI) independently on picture archival and communication system (PACS). Images at 1.5 and 3 T were reviewed by both at separate occasions with a gap of at least 4 weeks. Individual station images as well as whole-body stitched images of each WBMRI exam were reviewed to assess five aspects/categories: Vertebral column visibility, liver visibility, visibility of distal tibia/fibula, artifact grading, and overall image quality. These five aspects were analyzed using the grading system summarized in [Table 1] and [Table 2] and some are illustrated in [Figure 1] and [Figure 2]. Two regions of bone marrow space were selected – one that may be affected by breathing movement (vertebral column) and another that is unlikely to be affected by movement (distal tibia/fibula). For soft tissue visibility, the liver was selected, as it is a common site of metastases. This scoring system was developed by us. Before reviewing the actual images of the study group, both reviewers together reviewed 10 WBMRI exams (5 at 1.5 T and 5 at 3 T) outside the study group to ensure consensus regarding the scoring system used in this study. Both reviewers also listed the known artifacts for each exam. The pathologic findings were not analyzed in this study.
|Figure 1: (A-D): (A-D) Overall image quality comparison. (A) WBMRI image at 1.5 T shows excellent image quality rated as grade 4. (B) WBMRI image at 3 T shows good image quality rated as grade 3 that shows some movement artifacts affecting the lower legs and feet (arrows) and prominent chemical shift artifacts (arrowheads). (C) WBMRI image at 3 T shows fair image quality rated as grade 2 because of artifacts like ghosting (broken arrows), chemical shift (arrowheads), Moire fringe-like artifacts (arrows), and reduced visibility of liver and marrow in legs. (D) WBMRI image at 3 T shows poor image quality rated as grade 1/non-diagnostic because of extensive movement artifacts|
Click here to view
|Figure 2: (A-E): (A-E) Artifacts (arrows). (A) Breathing artifacts affecting the entire image. (B) Ghosting related to patient motion. (C) Pulsation artifacts in the feet from posterior tibial arteries. (D) Chemical shift at the interface of subcutaneous and muscle compartments in thighs. (E) Moire fringe-like artifacts at the body surface|
Click here to view
Inter-observer agreement between the two reviewers was calculated for all five categories of image analysis using Kendall's coefficient of Concordance (W). W-values less than 0.20 were considered as poor agreement, 0.21–0.40 as fair agreement, 0.41–0.60 as moderate agreement, 0.61–0.80 as substantial agreement, and greater than 0.81 as almost perfect agreement. Average of the mean scores given by the two readers for each category was used to compare 1.5 and 3 T using non-parametric Signed Rank test. Data analysis was performed using SAS 9.3. In all analyses, P values <0.05 were considered statistically significant.
| Results|| |
There was substantial agreement between the two reviewers for all five categories and both field strengths. W-values of marrow visibility, liver visibility, and overall image quality were higher for 3 T as compared to 1.5 T. W-values are summarized in [Table 3].
The mean scores for all five categories by both readers were higher for 1.5 T images [Table 4]. Comparison was done between the mean scores at 1.5 and 3 T averaged between the two readers. The difference between the averages of mean scores of the two field strengths was statistically significant [Table 5], indicating less artifact, better fixed structure visibility and overall image quality at 1.5 T as compared to 3 T.
|Table 5: Comparison of average of mean scores by two readers between 1.5 and 3 T|
Click here to view
Artifacts listed by both readers were summarized and compared for two field strengths. Motion-related artifacts were seen to be the most common on both sets of images, including patient movement, motion due to pulsation, and bowel peristalsis. One form or the other of motion-related artifacts was seen in all the examinations (100%) on both 1.5 and 3 T. Ghosting was present in six cases on 1.5 T (28%) and nine cases on 3 T (42%). Pulsation artifacts from vessels were seen in three cases (14%) on 1.5 T images as compared to those seen in nine cases (42%) on 3 T images. Chemical shift artifacts, predominantly affecting the interface between subcutaneous tissue and muscles in the extremities, were seen in 1 case (5%) on 1.5 T and in all 21 cases on 3 T (100%). Interference pattern seen as alternate curved bands of bright and dark signal at the periphery of the images, similar to Moire fringe artifacts resulting from field inhomogeneity, were seen in nine cases on 3 T (42%), but on none of the 1.5 T images. Susceptibility artifacts related to bowel and marrow were seen in six cases on 3 T (28%), but on none of the 1.5 T images.
| Discussion|| |
WBMRI is being used with increasing frequency in children for various oncologic and non-oncologic indications. It has been shown to be useful in the assessment of lymphoma. Even though this application, even in combination with DWI, is limited by its inability to differentiate between normal and abnormal lymph nodes, WBMRI has been shown to have high sensitivity and specificity for detection of malignant lymph nodes based on size criteria (short-axis diameter >1 cm). WBMRI is more sensitive than bone scan for detection of bone metastases in children.,, It is used for screening of children with cancer predisposition syndromes like Li-Fraumeni syndrome and in children with retinoblastoma for detection of metastases and osteosarcoma. Some non-oncologic applications of WBMRI in children include chronic recurrent multifocal osteomyelitis (CRMO), Langerhans cell histiocytosis, generalized osteonecrosis after cancer treatment, generalized vascular malformations like hemangiomatosis, lymphangiomatosis, and Klippel–Trenaunay syndrome, and fever of unknown origin. WBMRI in combination with DWI has the potential to replace PET/CT in the future. PET/MRI is likely to replace PET/CT in children because of lack of ionizing radiation from the CT component.
Various combinations of sequences are used for WBMRI. Coronal STIR sequence alone or in combination with other sequences is the most commonly used one for WBMRI. Most pathologic tissues are proton rich and have prolonged T1 and T2 relaxation times resulting in high signal intensity on STIR images. Robust and homogeneous fat suppression is another advantage of STIR sequence that is useful for evaluating bone marrow in children, most of whom have hypercellular marrow. The WBMRI at our institution includes only coronal STIR sequence covering the whole body. It is performed with a body coil integrated within the magnet bore, which in combination with sliding table platform obviates repositioning of patients at each station. Imaging with body coil has lower signal-to-noise ratio (SNR) as compared to phased-array coil. Despite this and other limitations like suboptimal depiction of sternum, ribs, scapula, and skull, and the lower sensitivity of coronal plane for detection of lymphadenopathy, coronal STIR serves the purpose well as a quick screening or "search" sequence for marrow and soft tissue abnormalities. With newer state-of-the-art systems with the capability to cover entire body with surface coil, potentially much better quality can be achieved.
MR signal improves with the magnetic field strength. MR imaging at 3 T provides high SNR, spatial resolution, and temporal resolution. Improved diagnostic accuracy and image quality has been reported with 3 T for imaging of brain, heart, vessels, musculoskeletal structures, and MR cholangiopancreatography., WBMRI at 3 T is expected to provide good-quality images in shorter time as compared to 1.5 T. However, these advantages are associated with increased motion-and pulsation-related artifacts, field inhomogeneity-related artifacts, increased susceptibility, increased chemical shift, and reduced T1 contrast due to longer T1 relaxation time at 3 T [Figure 3]. Increased chemical shift, increased susceptibility, and difficulty in achieving homogeneous magnetic field at 3 T as compared to 1.5 T result in greater artifacts at 3 T. Artifacts related to motion and susceptibility at 3 T can be minimized by use of parallel imaging where high signal at 3 T and parallel imaging act complementary to each other.
|Figure 3: (A and B): (A and B) Overall comparison of images at 1.5 and 3 T. WBMRI images at 1.5 T (A) and at 3 T (B) in a 10-year-old boy do not show any abnormality. The overall image quality was rated as 3 for both images. However, there are some inherent differences between the two images, including better lung visibility at 1.5 T, darker bones at 3 T (seen in proximal humeri, pelvic bones, and upper lumbar vertebrae), better signal in arms at 3 T, and greater chemical shift artifact at 3 T (arrows)|
Click here to view
The previous comparison study between 1.5 and 3 T WBMRI in adults by Schmidt et al. involved multiple sequences including coronal STIR. Even though overall image quality at 1.5 T was significantly better than at 3 T, it was rated as "good" on both field strengths in their study, suggesting feasibility of 3 T for WBMRI. The difference in the image quality of STIR between 1.5 and 3 T was due to significantly more artifacts affecting STIR at 3 T, including dielectric effects, pulsation artifacts, and image inhomogeneity. Similar to that study, our results showed significantly better overall image quality at 1.5 T and also suggested feasibility of 3 T for performance of WBMRI, with overall image quality rated at 2.74 for 3 T versus 3.10 for 1.5 T, both of which are within a good range. Most artifacts were seen with greater frequency at 3 T in our study, with some like chemical shift, susceptibility, and Moire fringe-like artifacts seen almost exclusively at 3 T.
Abdominal DWI performed with free breathing has been shown to be of better quality with less artifacts at 1.5 T as compared to 3 T. However, in the same study, image quality of DWI with breath hold and use of parallel imaging was better at 3 T as compared to 1.5 T, with similar artifacts scores. Our WBMRI coronal STIR sequence is performed with free breathing even in stations involving the chest and abdomen. This was reflected in significantly more artifact in these regions on our images. Moreover, we could not use parallel imaging because our WBMRI is performed with the body coil.
Since we could not compare assessment of pathology in this study, we tried to evaluate fixed structure visibility, which in our opinion indirectly reflects diagnostic ability of WBMRI at 1.5 and 3 T. Two regions of bone marrow space were selected – one that may be affected by breathing movement (vertebral column) and another that is unlikely to be affected by movement (distal tibia/fibula). For soft tissue visibility, the liver was selected, as it is a common site of metastases. Visibility of all three regions was significantly better at 1.5 T than at 3 T. However, similar to overall image quality, fixed structure visibility for both field strengths was within a good range [Table 5]. Higher visibility scores at 1.5 T for liver and vertebral column were mainly due to fewer artifacts related to breathing and other motion-related artifacts, while higher scores for marrow visibility in distal tibia and fibula at 1.5 T were due to less inhomogeneity and darkening as compared to 3 T. Higher inter-observer agreement (W-values) for scores at 3 T may be related to more severe degree of artifacts and relatively less visibility of fixed structures that are usually more obvious.
Our study has a few limitations. It is a retrospective study with a small sample size. The study period falls between 2008 and 2011. In the last few years since then, MRI technology has improved and some of the artifacts may have been reduced and image quality has improved. It does not directly compare assessment of pathology on two field strengths or impact of image quality and artifacts on lesion detection. Nonetheless, it provides comparative data on image quality and artifacts on the most commonly used sequence for WBMRI at two most commonly used field strengths in children.
| Conclusion|| |
Similar to previous studies in adults, WBMRI performed with coronal STIR sequence at 1.5 T has significantly better image quality, fixed structure visibility and fewer artifacts, as compared to WBMRI at 3 T in children. This difference is unlikely to significantly affect detection of pathology on 3 T WBMRI, as the image quality score at 3 T was also within a good range. Large studies comparing actual detection of pathology at two field strengths are required.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Chavhan GB, Babyn PS. Whole-body MR imaging in children: Principles, technique, current applications, and future directions. Radiographics 2011;31:1757-72.
Monsalve J, Kapur J, Malkin D, Babyn PS. Imaging of cancer predisposition syndromes in children. Radiographics 2011;31:263-80.
Fritz J, Tzaribatchev N, Claussen CD, Carrino JA, Horger MS. Chronic recurrent multifocal osteomyelitis: Comparison of whole-body MR imaging with radiography and correlation with clinical and laboratory data. Radiology 2009;252:842-51.
Goo HW, Yang DH, Ra YS, Song JS, Im HJ, Seo JJ, et al.
Whole-body MRI of Langerhans cell histiocytosis: Comparison with radiography and bone scintigraphy. Pediatr Radiol 2006;36:1019-31.
Laffan EE, O'Connor R, Ryan SP, Donoghue VB. Whole-body magnetic resonance imaging: A useful additional sequence in paediatric imaging. Pediatr Radiol 2004;34:472-80.
Kellenberger CJ, Miller SF, Khan M, Gilday DL, Weitzman S, Babyn PS. Initial experience with FSE STIR whole-body MR imaging for staging lymphoma in children. Eur Radiol 2004;14:1829-41.
Kellenberger CJ, Epelman M, Miller SF, Babyn PS. Fast STIR whole-body MR imaging in children. Radiographics 2004;24:1317-30.
Kembhavi SA, Rangarajan V, Shah S, Qureshi S, Arora B, Juvekar S, et al.
Prospective observational study on diagnostic accuracy of whole-body MRI in solid small round cell tumours. Clin Radiol 2014;69:900-8.
Bannas P, Finck-Wedel AK, Buhk JH, Bley TA, Koops A, Kooijman H, et al.
Comparison of whole body MR angiography at 1.5 and 3 Tesla in patients with hereditary hyperlipidemia. Acta Radiol 2011;52:547-53.
Thieme SF, Dietrich O, Maxien D, Nikolaou K, Schoenberg SO, Reiser M, et al.
Oxygen-enhanced MRI of the lungs: Intraindividual comparison between 1.5 and 3 Tesla. Rofo 2011;183:358-64.
Yang PC, Nguyen P, Shimakawa A, Brittain J, Pauly J, Nishimura D, et al.
Spiral magnetic resonance coronary angiography – Direct comparison of 1.5 Tesla vs 3 Tesla. J Cardiovasc Magn Reson 2004;6:877-84.
Hinton DP, Wald LL, Pitts J, Schmitt F. Comparison of cardiac MRI on 1.5 and 3.0 Tesla clinical whole body systems. Invest Radiol 2003;38:436-42.
Saremi F, Jalili M, Sefidbakht S, Channual S, Quane L, Naderi N, et al.
Diffusion-weighted imaging of the abdomen at 3 T: Image quality comparison with 1.5-T magnet using 3 different imaging sequences. J Comput Assist Tomogr 2011;35:317-25.
Schmidt GP, Wintersperger B, Graser A, Baur-Melnyk A, Reiser MF, Schoenberg SO. High-resolution whole-body magnetic resonance imaging applications at 1.5 and 3 Tesla: A comparative study. Invest Radiol 2007;42:449-59.
Punwani S, Taylor SA, Bainbridge A, Prakash V, Bandula S, De Vita E, et al.
Pediatric and adolescent lymphoma: Comparison of whole-body STIR half-Fourier RARE MR imaging with an enhanced PET/CT reference for initial staging. Radiology 2010;255:182-90.
Mazumdar A, Siegel MJ, Narra V, Luchtman-Jones L. Whole-body fast inversion recovery MR imaging of small cell neoplasms in pediatric patients: A pilot study. AJR Am J Roentgenol 2002;179:1261-6.
Kumar J, Seith A, Kumar A, Sharma R, Bakhshi S, Kumar R, et al.
Whole-body MR imaging with the use of parallel imaging for detection of skeletal metastases in pediatric patients with small-cell neoplasms: Comparison with skeletal scintigraphy and FDG PET/CT. Pediatr Radiol 2008;38:953-62.
Goo HW, Choi SH, Ghim T, Moon HN, Seo JJ. Whole-body MRI of paediatric malignant tumours: Comparison with conventional oncological imaging methods. Pediatr Radiol 2005;35:766-73.
Darge K, Jaramillo D, Siegel MJ. Whole-body MRI in children: Current status and future applications. Eur J Radiol 2008;68:289-98.
Hirsch W, Krohmer S, Kluge R, Krausse A, Sorge I. Preliminary results in whole-body MRI in children – A prospective study [abstr]. Pediatr Radiol 2005;35 Issue 2-Suppl:S89.
Chavhan GB, Babyn PS, Singh M, Vidarsson L, Shroff M. MR imaging at 3.0 T in children: Technical differences, safety issues, and initial experience. Radiographics 2009;29:1451-66.
Govind B Chavhan
Department of Diagnostic Imaging, The Hospital for Sick Children and Medical Imaging Department, University of Toronto, 555 University Avenue, Toronto, Ontario M5G 1X8
Source of Support: None, Conflict of Interest: None
[Figure 1], [Figure 2], [Figure 3]
[Table 1], [Table 2], [Table 3], [Table 4], [Table 5]